High cost-effective precise colorimeter NR60CP 3nh with delta E 0.03
There are many terms used to describe color. Here are some basic definitions to help clarify:
Hue: This term is what we refer to often as colors: red, orange, yellow, etc. Hues are differentiated by different wavelengths on the color spectrum.
Value: The brightness of a color, or how much light it emits.
Tint, Shade and Tone: Tinting a color means adding white to the hue, shading means adding black to a hue, and toning means to add grey to a hue.
Saturation: A color of low saturation is closer to neutral than to the pure hue, whereas a highly
saturated color is further from neutral and more richly pigmented. Colors can be muted or neutralized by combining them with varying amounts of their color-wheel opposite.
Cool/Warm colors: Cool colors are the range of hues surrounding blue, including green and purple. Warm colors are colors surrounding orange, including red and yellow. Reddish-purple and yellow-green are points on the color wheel where cool and warm intersect. For skin tones, cool is typically a more blue or blue combined with pink, whereas warm appears more yellow or golden.
High cost-effective precise colorimeter NR60CP 3nh with delta E 0.03 features high accuracy, powerful functions, and high cost-effective to repay the support and trust of our customers. Colorimeter NR60CP 3nh with delta E 0.03 is widely used in plastic, electronic, paint, ink, textile, garment, printing and dyeing, food, medical, cosmetic, industries, scientific research institutes, schools and laboratories.
|Illuminating/Viewing Geometry||8° / d; CI E No. 15,GB/ T 3978|
|Light Source||LED sources|
|Sensor||Silicon photodiode array|
|Measuring Aperture||Φ8mm large area end-face; Φ4mm small area concave-convex end-face|
|Color Space||CI E L AB,XYZ,LCh,CI E RGB,CI E LUV|
|Other Chromaticity Data||WI ( E313,CI E, AATCC,Hunt er ),YI ( D1925, 313),Color Fastness, Staining Fastness, J PC79,BFD( 1. 5: 1),FMCI I|
|Observer||CI E 10°|
|Display Data||Colorimetric Value, Color Difference Value/Graph, PASS/FAIL Result, Color Offset,|
|Measurement Time||0. 4s|
|Repeatability||ΔE* ab 0. 03(Average of 30 measurement of standard white plate within 5s)|
|Errors between each instrument||within Delta E*ab 0.2|
|Battery||rechargeable lithium-ion battery 3. 7V @ 3200mAh|
|Lamp Life||5 years, more than 1.6 million measurements|
|Display Screen||TFT 2.8 inch(16: 9)|
|Data Memory||100 Standards, 20000 Samples|
|Optional Accessory||miniature thermal printer, powder test box|