Eider GranadaThe product and the service was excelent. it came to colombia without many problems (just aduana problems)..supplier 100% recomenmed
Josiah HallEverything was delivered in perfect condition. After assembly, I confirmed operation. Thank you for this product.
Mihai PaunescuEverything went as expected and the product arrived in good condition. 3NH is a good supplier and I will definitely use them in the future.
NS 810 3nh Spectrophotometer D/8 Optical Plastic Industry Color Difference Check
Contact me for free samples and coupons.
Wechat: 0086 18588475571
If you have any concern, we provide 24-hour online help.x
|Illumination/Observation System||D/8 Structure||Wavelength Range||400-700nm|
|Wavelength Interval||10nm||Color Difference Formula||ΔE*ab,ΔE*uv,ΔE*94,ΔE*cmc(2:1),ΔE*cmc(1:1),ΔE*00 V|
|Chromaticity Data||WI,YI,Mt,color Strength,color Stain,color Fastness,opacity||Observer||2-degree/10-degree|
|Repeatability||Within Delta E*ab 0.04||Illuminant||D65,A,C,D50,D55,D75,F1,F2,F3,F4,F5,F6,F7,F8,F9,F10,F11,F12|
Plastic Industry 3nh Spectrophotometer,
Color Difference Check 3nh Spectrophotometer,
700nm Colour Measurement Instruments
3nh High Quality Spectrophotometer NS 810 Focus on color for Plastic Industry Color Difference Check
Based on large integrating sphere theory, NS810 portable spectrophotometer adopts d/8 optical structure with 8mm aperture measurement. It is much more precise and stable than any other colorimeter model.
Our 3nh brand spectrophotometer colorimeter can specify using a universla color language and ensure suppliers are quality controlled and color managed. Plus, our SQC8 software for free make controlling color fast and easy.
NS810 spectrophotometer is widely used in plastic, electronic, paint, ink, textile, garment, printing and dyeing, food, medical, cosmetic, industries, scientific research institutes, schools and laboratories. It can measure reflectance spectrum and other color index precisely.
|Illumination/observation system||d/8 (diffused illumination, 8-degree viewing angle) Conforms to CIE No.15, GB/T 3978.|
|Integrating sphere Size||Φ58mm|
|Light Source||Combined LED sources|
|Sensor||Silicon photodiode array|
|Color Space||CIE LAB, XYZ, Yxy, LCh, CIE LUV,LAB&WI&YI|
|Color difference Formula||ΔE*ab,ΔE*uv,ΔE*94,ΔE*cmc(2:1),ΔE*cmc(1:1),ΔE*cmc(l:c),CIE2000ΔE*00,ΔE(h)|
|Other Chromaticity Data||
WI(ASTM E313,CIE/ISO,AATCC, Hunter),
YI(ASTM D1925,ASTM 313),
Metamerism Index (Mt), Color Strength
Color Stain, Color Fastness
|Display Data||Spectral Value/Graph, Colorimetric Value, Color Difference Value/Graph, PASS/FAIL Result, Color Offset, Color Simulation, color index setting(ΔE*94, ΔE*cmc, ΔE2000), tolerance prompt, reverse prompt, time setting, language setting, restore factory setting|
|Repeatability||Spectral Reflectance: standard deviation within 0.1%(400~700nm: within 0.2%)
Colorimetric Value: Standard deviation within DeltaE*ab 0.04 (Measurement conditions: white calibration plate measured 30 times at 5 seconds intervals after white calibration was performed.)
|Inter Instrument Agreement||Within DeltaE*ab 0.2 (Average for 12 BCRA Series II color tiles)|
|Battery||Li-ion battery. 2800 times within 8 hours.|
|Lamp Life||5 years, more than 1.6 million measurements|
|Display Screen||TFT 3.5inch, Capacitive Touch Screen|
|Data Memory||1000 Standards, 15000 Samples|
|Standard Accessory||Power Adapter, Li-ion Battery, Operating Instruction, CD-ROM (containing management software), Data Line, White and Black Calibration Cavity, Protective Cover and Wrist Strap|
|Optional Accessory||Universal Test Components, Micro Printer, Powder Test Box|
|Notes||The specifications are subject to change without notice.|
Related Knowledge about Spectrophotometer:
-L(Lightness)-0 is black,100 is white
-a(Red-Green)-positive values are red; negative values are green and 0 is neutral
-b(Blue-Yellow)-positive values are yellow;negative values are blue abd 0 is neutral
L: Lightness,value is the same in each scale C:Chroma H:Hue Angle
dE Total Color difference
Delta-E is a measurement used to indicate how much a color deviates from an accepted standards.The higher the dE, the more inaccurate the color.